# Advanced Energy OLS10KDC Rack-Mount 电源 (1kV ~ 100kV/10kW)

制造商: Advanced Energy Model: OLS10KDC 保障: 12 Month - 联系

- Output Power: 10 kW for the HE (High Energy) supply and 1 kW for the LE (Low Energy) supply at full rated output voltage and current

- Output Voltage: The OLS10KD offers maximum HE output voltages from 10 kV ~ 100 kV, and LE output voltages from 1kV ~ 10 kV

- Output Current: Up to 1 A for 10 kV HE and 1kV LE, and and 100 mA for 100 kV HE and 10 kV LE

- Input Voltage: 380 VAC ±10% (342 VAC ~ 418 VAC) 47~63 Hz 3 phase plus protective earth

- Input Current: Less than 36 A per phase

- Specification Range: Specifications apply above 5% of rated output voltage for both HE and LE supplies. The output can be controlled down to less than 0.25% of rated output voltage.

- Calibration: Calibration between Voltage Demand, Output Voltage and Voltage Monitor ±0.2% of setting or ±0.05% of rating, whichever is greater

- Recovery Time: Less than 200 ms to within 0.1% of previous operating level following a short circuit or arc. Maximum overshoot 2% of rated output voltage

- Temperature Coefficient: Less than 100 ppm/°C

- Drift: Less than 0.01% per hour after 1 hour’s warm up, typically less than 0.02% per 8 hours after 1 hour’s warm up, at constant load, line and temperature

- Operating Temperature: 0°C ~ +40°C

- Storage Temperature: -20°C ~ +70°C (-4°F ~ 158°F)

- Humidity: 80% maximum relative humidity up to 31°C, reducing linearly to 50% at 40°C. Non-condensing (ref BS EN61010-1)

- Altitude: Sea level up to 2000 metres (6500 feet)

- Usage: Indoor use only

- Voltage Ripple (High and Low Range):

Voltage Mode: Less than 0.05% of rated voltage +2 V, peak to peak or less than 0.01% of rated voltage +1 V, rms

Current Mode:Less than 0.5% of rated voltage peak to peak or less than 0.1% of rated voltage rms

- Voltage Regulation (High and Low Range):

Line: Less than 0.05% +0.5 V change in output voltage for a 10% change in line voltage

Load: Less than 0.05% +0.5 V change in output voltage for 0 to 100% change in load current

- Current Regulation (High and Low Range):

Line: Less than 0.5% of rated current for a ±10% change in line voltage

Load: Less than 0.5% of rated current for 0 ~ 100% change in output voltage

- Dimensions (W x D x H):

Models Up to 80 kV: 43 x 51.6 x 26.6 cm3 (17 x 20.3 x 10.5 in3)

Models Over 80 kV: 43 x 61.4 x 26.6 cm3 (17 x 24.2 x 10.5 in3)

- Weight: 45 kg (99 lb)

- Interface:

Remote Control Interface: Analog, control is via a 25-way female D-type connector

Optional digital RS232 remote control, 9-way female D-type connector available

- Options & Accessories

Positive or negative polarity

Front panel and control options

HV output cables available — 3 m, 5 m, or 10 m

- 质量承诺
- 正品保修
- 送货到家
- 交易简单化